The collection needle tube drops and impedes arm travel, or prevents correct collection, what should I do?
Concerned instruments: puriFlash 215, puriFlash 430, puriFlash 450, puriFlash 4100, puriFlash 4250
Concerned module: collector
The ferrule holding the tubing in place is incorrectly positioned.
Simply reposition the ferrule above the needle such that the tubing can no longer fall.